Width: 8.6 in (21.9 cm) without temperature, 10.0 in (25.4 cm) with temperature
Depth: 5.3 in (13.5 cm)
Weight: 5.4 lb (2.4 kg) including battery
Mountings: Self-supporting on rubber feet or pole mounted
Portability: Carried by recessed handle
Power converter universal P/N: 2018859-001
Protection against electrical shock: Class II
AC input Voltage: 100 to 250VAC, 12VA
DC output Voltage: 12VDC at 1A
The AC mains power adapter contains a nonresettable and nonreplaceable fuse.
Protection against electrical shock: Internally powered or Class II when powered from specified external power supply.
DC input voltage: 12 VDC, supplied from a source conforming to IEC 60601-1.
Fuses: The monitor contains three fuses. The fuses are mounted within the monitor. The fuses protect the low voltage DC input, the battery, and the remote alarm output. The +5 V output on the host port connector is regulated by internal supply.
Battery: Capacity: 6V 3.3 Ahr sealed lead acid battery protected by internal auto-resetting fuse and thermal protection Battery Life: 8.1 hours with a usage scenario of: NIBP determinations every 15 minutes with SpO2 and temperature active. 11.5 hours non-SpO2 versions with a usage scenario of: NIBP determinations every 15 minutes with temperature active.
Charge time: Approximately 5 hours from full discharge when the monitor is off. Approximately 8 hours when the monitor on.
Operating Temperature : + 5°C to + 40°C (+ 41°F to + 104°F)
Operating Atmospheric Pressure: 700 hPa to 1060 hPa
Storage / Transportation
Storage Temperature &ndash 20°C to + 50°C (&ndash 4°F to + 122°F)
Atmospheric Pressure: 500 hPa to 1060 hPa
Humidity Range: 5% to 95% noncondensing
Radio Frequency: Complies with IEC Publication 60601-1-2 (2001). Medical Electrical Equipment, Electromagnetic Compatibility Requirements and Tests and CISPR 11 (Group 1, Class B) for radiated and conducted emissions
Send Enquiry for
GE Medical CARESCAPE V100 monitor